http://www.loc.gov/mods/v3 http://www.loc.gov/standards/mods/v3/mods-3-2.xsd
Response comparison of an optically stimulated luminescent dosimeter, a direct-ion storage dosimeter, and a thermoluminescence dosimeter
This study was undertaken to compare the response of three dosimeters to different environments. Comanche Peak Nuclear Power Plant wants to replace the current badge of record. The RaDos DIS-1 direct-ion storage dosimeter (DIS-1) and the Landauer InLight optically stimulated luminescence dosimeter (OSL) are the two candidates for replacement of the Panasonic UD-802 thermoluminescence dosimeter (TLD). The dosimeters were compared in five categories: dose linearity, dose-rate linearity, fade response, humidity response, and the angular dependence of the dosimeters. The major results include verified linear relationship evidence for dose and doserate and a better fade response for both the DIS-1 and OSL. The TLDs faded by 9.2% over a month and the DIS-1 and OSL faded by 4.2% and 1%, respectively. Following a dose of 557.5 mrem, the dosimeters were exposed to different relative humidites. The dose to the DIS-1 and OSL did not change drastically while the TLDs dose readout was reduced by 10%. Finally, the angular dependence of the dosimeters was compared and the worst responses were 66% at 90[degrees] in the horizontal orientation for the OSL and 1.7% at 90[degrees] in the horizontal orientation for the DIS-1. Based on the results of these tests the OSL seems like a more viable candidate for the new badge or record.
Title from author supplied metadata (automated record created 2010-03-12 12:08:51).
Master of Science Texas A&M University 2008 http://hdl.handle.net/1969.1/ETD-TAMU-2979